Definitions

# galvanometer

[gal-vuh-nom-i-ter]
galvanometer, instrument used to determine the presence, direction, and strength of an electric current in a conductor. All galvanometers are based upon the discovery by Hans C. Oersted that a magnetic needle is deflected by the presence of an electric current in a nearby conductor. When an electric current is passing through the conductor, the magnetic needle tends to turn at right angles to the conductor so that its direction is parallel to the lines of induction around the conductor and its north pole points in the direction in which these lines of induction flow. In general, the extent to which the needle turns is dependent upon the strength of the current. In the first galvanometers, a freely turning magnetic needle was hung in a coil of wire; in later versions the magnet was fixed and the coil made movable. Modern galvanometers are of this movable-coil type and are called d'Arsonval galvanometers (after Arsène d'Arsonval, a French physicist). If a pointer is attached to the moving coil so that it passes over a suitably calibrated scale, the galvanometer can be used to measure quantitatively the current passing through it. Such calibrated galvanometers are used in many electrical measuring devices. The DC ammeter, an instrument for measuring direct current, often consists of a calibrated galvanometer through which the current to be measured is made to pass. Since heavy currents would damage the galvanometer, a bypass, or shunt, is provided so that only a certain known percentage of the current passes through the galvanometer. By measuring the known percentage of the current, one arrives at the total current. The DC voltmeter, which can measure direct voltage, consists of a calibrated galvanometer connected in series (see electric circuit) with a high resistance. To measure the voltage between two points, one connects the voltmeter between them. The current through the galvanometer (and hence the pointer reading) is then proportional to the voltage (see Ohm's law).

Instrument for measuring small electric currents by deflection of a moving coil. A common galvanometer consists of a light coil of wire suspended from a metallic ribbon between the poles of a permanent magnet. As current passes through the coil, the magnetic field it produces reacts with the magnetic field of the permanent magnet, producing a torque. The torque causes the coil to rotate, moving an attached needle or mirror. The angle of rotation, which provides a measure of the current flowing in the coil, is measured by the movement of the needle or by the deflection of a beam of light reflected from the mirror.

A galvanometer is a type of ammeter; an instrument for detecting and measuring electric current. It is an analog electromechanical transducer that produces a rotary deflection, through a limited arc, in response to electric current flowing through its coil. The term has expanded to include uses of the same mechanism in recording, positioning, and servomechanism equipment.

## History

Deflection of a magnetic compass needle by current in a wire was first described by Hans Oersted in 1820. The phenomenon was studied both for its own sake and as a means of measuring electrical current. The earliest galvanometer was reported by Johann (Johan) Schweigger of Nuremberg at the University of Halle on 16 September 1820. André-Marie Ampère also contributed to its development. Early designs increased the effect of the magnetic field due to the current by using multiple turns of wire; the instruments were at first called "multipliers" due to this common design feature. The term "galvanometer", in common use by 1836, derives from the surname of Italian electricity researcher Luigi Galvani, who discovered that electric current could make a frog's leg jerk.

Originally the instruments relied on the Earth's magnetic field to provide the restoring force for the compass needle; these were called "tangent" galvanometers and had to be oriented before use. Later instruments of the "astatic" type used opposing magnets to become independent of the Earth's field and would operate in any orientation. The most sensitive form, the Thompson or mirror galvanometer, was invented by William Thomson (Lord Kelvin). Instead of a compass needle, it used tiny magnets attached to a small lightweight mirror, suspended by a thread; the deflection of a beam of light greatly magnified the deflection due to small currents. Alternatively the deflection of the suspended magnets could be observed directly through a microscope.

The ability to quantitatively measure voltage and current allowed Georg Ohm to formulate Ohm's Law, which states that the voltage across an element is directly proportional to the current through it.

The early moving-magnet form of galvanometer had the disadvantage that it was affected by any magnets or iron masses near it, and its deflection was not linearly proportional to the current. In 1882 Jacques-Arsène d'Arsonval developed a form with a stationary permanent magnet and a moving coil of wire, suspended by coiled hair springs. The concentrated magnetic field and delicate suspension made these instruments sensitive and they could be mounted in any position. By 1888 Edward Weston had brought out a commercial form of this instrument, which became a standard component in electrical equipment. This design is almost universally used in moving-vane meters today.

## Operation

The most familiar use is as an analog measuring instrument, often called a meter. It is used to measure the direct current (flow of electric charges) through an electric circuit. The D'Arsonval/Weston form used today is constructed with a small pivoting coil of wire in the field of a permanent magnet. The coil is attached to a thin pointer that traverses a calibrated scale. A tiny torsion spring pulls the coil and pointer to the zero position.

When a direct current (DC) flows through the coil, the coil generates a magnetic field. This field acts against the permanent magnet. The coil twists, pushing against the spring, and moves the pointer. The hand points at a scale indicating the electric current. Careful design of the pole pieces ensures that the magnetic field is uniform, so that the angular deflection of the pointer is proportional to the current. A useful meter generally contains provision for damping the mechanical resonance of the moving coil and pointer, so that the pointer settles quickly to its position without oscillation.

The basic sensitivity of a meter might be, for instance, 100 microamperes full scale (with a voltage drop of, say, 50 millivolts at full current). Such meters are often calibrated to read some other quantity that can be converted to a current of that magnitude. The use of current dividers, often called shunts, allows a meter to be calibrated to measure larger currents. A meter can be calibrated as a DC voltmeter if the resistance of the coil is known by calculating the voltage required to generate a full scale current. A meter can be configured to read other voltages by putting it in a voltage divider circuit. This is generally done by placing a resistor in series with the meter coil. A meter can be used to read resistance by placing it in series with a known voltage (a battery) and an adjustable resistor. In a preparatory step, the circuit is completed and the resistor adjusted to produce full scale deflection. When an unknown resistor is placed in series in the circuit the current will be less than full scale and an appropriately calibrated scale can display the value of the previously-unknown resistor.

Because the pointer of the meter is usually a small distance above the scale of the meter, parallax error can occur when the operator attempts to read the scale line that "lines up" with the pointer. To counter this, some meters include a mirror along the markings of the principal scale. The accuracy of the reading from a mirrored scale is improved by positioning one's head while reading the scale so that the pointer and the reflection of the pointer are aligned; at this point, the operator's eye must be directly above the pointer and any parallax error has been minimized.

## Types

Extremely sensitive measuring equipment once used mirror galvanometers that substituted a mirror for the pointer. A beam of light reflected from the mirror acted as a long, massless pointer. Such instruments were used as receivers for early trans-Atlantic telegraph systems, for instance. The moving beam of light could also be used to make a record on a moving photographic film, producing a graph of current versus time, in a device called an oscillograph.

Galvanometer mechanisms are used to position the pens of analog chart recorders such as used for making an electrocardiogram. Strip chart recorders with galvanometer driven pens might have a full scale frequency response of 100 Hz and several centimeters deflection. In some cases (the classical polygraph of movies or the electroencephalograph), the galvanometer is strong enough to move the pen while it remains in contact with the paper; the writing mechanism may be a heated tip on the needle writing on heat-sensitive paper or a fluid-fed pen. In other cases (the Rustrak recorders), the needle is only intermittently pressed against the writing medium; at that moment, an impression is made and then the pressure is removed, allowing the needle to move to a new position and the cycle repeats. In this case, the galvanometer need not be especially strong.

### Tangent galvanometer

A tangent galvanometer is an early measuring instrument used for the measurement of electric current. It works by using a compass needle to compare a magnetic field generated by the unknown current to the magnetic field of the Earth. It gets its name from its operating principle, the tangent law of magnetism, which states that the tangent of the angle a compass needle makes is proportional to the ratio of the strengths of the two perpendicular magnetic fields. It was first described by Claude Servais Mathias Pouillet in 1837.

A tangent galvanometer consists of a coil of insulated copper wire wound on a circular non-magnetic frame. The frame is mounted vertically on a horizontal base provided with levelling screws. The coil can be rotated on a vertical axis passing through its centre. A compass box is mounted horizontally at the centre of a circular scale. It consists of a tiny, powerful magnetic needle pivoted at the centre of the coil. The magnetic needle is free to rotate in the horizontal plane. The circular scale is divided into four quadrants. Each quadrant is graduated from 0° to 90°. A long thin aluminium pointer is attached to the needle at its centre and at right angle to it. To avoid errors due to parallax a plane mirror is mounted below the compass needle.

In operation, the instrument is first rotated until the magnetic field of the Earth, indicated by the compass needle, is parallel with the plane of the coil. Then the unknown current is applied to the coil. This creates a second magnetic field on the axis of the coil, perpendicular to the Earth's magnetic field. The compass needle responds to the vector sum of the two fields, and deflects to an angle equal to the tangent of the ratio of the two fields. From the angle read from the compass's scale, the current could be found from a table.

The current supply wires have to be wound in a small helix, like a pig's tail, otherwise the field due to the wire will affect the compass needle and an incorrect reading will be obtained.

#### Theory

When current is passed through the tangent galvanometer a magnetic field is created at its corners given by $B=\left\{mu_0 nIover 2r\right\}$ where I is the current in ampere, n is the number of turns of the coil and r is the radius of the coil.

If the galvanometer is set such that the plane of the coil is along the magnetic meridian i.e., B is perpendicular to $B_H$ ($B_H$ is the horizontal component of the Earth's magnetic field), the needle rests along the resultant. From tangent law, $B = B_H tantheta$, i.e.

$\left\{mu_0 nIover 2r\right\} = B_H tantheta$
or
$I=left\left(frac\left\{2rB_H\right\}\left\{mu_0 n\right\}right\right)tantheta$
or $I=K tantheta$, where K is called the Reduction Factor of the tangent galvanometer.

The value of $theta$ is taken at 45 degrees for maximum accuracy.

#### Geomagnetic field measurement

A tangent galvanometer can also be used to measure the magnitude of the horizontal component of the geomagnetic field. When used in this way, a low-voltage power source, such as a battery, is connected in series with a rheostat, the galvanometer, and an ammeter. The galvanometer is first aligned so that the coil is parallel to the geomagnetic field, whose direction is indicated by the compass when there is no current through the coils. The battery is then connected and the rheostat is adjusted until the compass needle deflects 45 degrees from the geomagnetic field, indicating that the magnitude of the magnetic field at the center of the coil is the same as that of the horizontal component of the geomagnetic field. This field strength can be calculated from the current as measured by the ammeter, the number of turns of the coil, and the radius of the coils.

## Uses

A major early use for galvanometers was for finding faults in telecommunications cables. They were superseded in this application late in the 20th century by time-domain reflectometers.

Since the 1980s, galvanometer-type analog meter movements may be displaced by analog to digital converters (ADCs) for some uses. A digital panel meter (DPM) contains an analog to digital converter and numeric display. The advantages of a digital instrument are higher precision and accuracy, but factors such as power consumption or cost may still favor application of analog meter movements.

Most modern uses for the galvanometer mechanism are in positioning and control systems. These are used in laser marking and projection, and in imaging application such as Optical Coherence Tomography (OCT) retinal scanning. Mirror galvanometer systems are used as beam positioning elements in laser optical systems. These are typically high power galvanometer mechanisms used with closed loop servo control systems. The newest generation of galvanometers designed for beam steering applications can have frequency responses over 10 kHz with appropriate servo technology. Examples of manufacturers of such systems are Cambridge Technology Inc. (www.camtech.com) and General Scanning (www.gsig.com).

A galvanometer appeared in an episode of the television medical drama House to function as an electrocardiogram for a patient whose severe and extensive burns prevented use of the normal electrodes.