There are a number of different "detection limits" that are commonly used. These include the instrument detection limit (IDL), the method detection limit (MDL) and the limit of quantitation (LOQ). Even when the same terminology is used, there can be differences in the LOD according to nuances of what definition is used and what type of noise contributes to the measurement and calibration.
The figure illustrates the relationship between the blank, the limit of detection (LOD), and the limit of quantification (LOQ) by showing the probability density function for normally distributed measurents at the blank, at the LOD defined as 3 * standard deviation of the blank, and at the LOQ defined as 10 * standard deviation of the blank. For a signal at the LOD, the alpha error (probability of false positive) is small (1%). However, the beta error (probability of a false negative) is 50% for a sample that has a concentration at the LOD (red line). This means a sample could contain an impurity at the LOD, but there is a 50% chance that a measurement would give a result less than the LOD. At the LOQ (blue line), there is minimal chance of a false negative.
Most analytical instruments produce a signal even when a blank (matrix without analyte) is analyzed. This signal is referred to as the noise level. The IDL is the analyte concentration that is required to produce a signal greater than three times the standard deviation of the noise level. This may be practically measured by analyzing 8 or more standards at the estimated IDL then calculating the standard deviation from the measured concentrations of those standards.
Many times there is more to the analytical method than just doing a reaction or submitting it to direct analysis. For example it might be necessary to heat a sample that is to be analyzed for a particular metal with the addition of acid first (this is called digestion). The sample may also be diluted or concentrated prior to analysis on an instrument. Additional steps in an analysis add additional opportunities for error. Since detection limits are defined in terms of error, this will naturally increase the measured detection limit. This detection limit (with all steps of the analysis included) is called the MDL. The practical method for determining the MDL is to analyze 7 samples of concentration near the expected limit of detection. The standard deviation is then determined. The one-sided t distribution is determined and multiplied versus the determined standard deviation. For seven samples (with six degrees of freedom) the t value for a 99% confidence interval is 3.14. Rather than performing the complete analysis of seven identical samples, if the Instrument Detection Limit is known, the MDL may be estimated by multiplying the Instrument Detection Limit or Lower Level of Detection by the dilution prior to analyzing the sample solution on the instrument. This estimation, however, ignores any uncertainty that arises from performing the sample preparation and will therefore probably underestimate the true MDL.
Just because we can tell something from noise does not mean that we can necessarily know how much of the material there actually is. The LOQ is the limit at which we can reasonably tell the difference between two different values. The LOQ is drastically different between labs so another detection limit is commonly used that is referred to as the Practical Quantitation Limit (PQL). The PQL is defined simply as about 5 times the MDL.
Suppose you are at an airport with lots of noise from jets taking off. If the person next to you speaks softly, you will probably not hear them. Their voice is less than the LOD. If they speak a bit louder, you may hear them but it is not possible to be certain of what they are saying and there is still a good chance you may not hear them. Their voice is >LOD but
Comparison of the lower confidence limit to the fixed-percentage method for assessing airway obstruction in routine clinical practice.(Report)
Nov 01, 2011; Introduction Demonstration of a reduced ratio of [FEV.sub.1] to VC (vital capacity) or FVC (forced vital capacity) on spirometry...
Research from George Washington University, Biostatistics Center in the area of clinical trial research published.(Report)
Mar 15, 2010; Research findings, 'Futility interim monitoring with control of type I and II error probabilities using the interim Z-value or...